Heated Static Headspace Injector
- Uses standard 40 mL VOA vials
- VOA vial sleeve thermostatted from ambient to 90°C
- Gas sampling valve with fixed volume loop
- Gauge displays actual vial pressure
- Liquid, solid, or powder sample
- Complete PeakSimple control
The Heated Static Headspace injector is useful for the analysis of volatiles, especially where the sample matrix is dirty. A 10-port gas sampling valve and fixed volume sample loop are used for maximum precision.
The thermostatted headspace sleeve accepts standard 40 mL VOA vials containing 10 - 20 mL of sample.
As the vial is inserted into the headspace sleeve, two needles puncture the septum top of the vial. Purge gas enters through one needle to pressurize the vial, and the other needle carries headspace vapors to the loop of the gas sampling valve. A solenoid valve located at the loop exit is opened under PeakSimple data system control to allow headspace vapors to purge through the loop just prior to injecting the loop contents onto the column. The entire headspace sleeve is mechanically agitated under control of the data system.
The headpsace sleeve is thermostatted from ambient to 90°C under PeakSimple control, and can be cooled down before removing the VOA vial.
Share this Product
More from this collection
Questions & Answers
Have a Question?
Be the first to ask a question about this.